Electrophoresis power supplies are commonly found in biology and other life sciences laboratories. These power supplies are usually capable of supplying high voltages and high currents required for gel electrophoresis–a method used for separating DNA, RNA and other protein fragments based on their size and charge. There are many used electrophoresis power supplies out there in the second hand market and can be bought quite cheaply. I am curious whether these electrophoresis power supplies are suitable for electronics lab use as a lab grade high voltage power supply can be quite expensive. So I recently picked up one from eBay to take a look.
The MMI 5300 was a memory chip from the early 1970s, storing 1024 bits in tiny fuses.1 Unlike regular RAM chips, this was a PROM (Programmable Read-Only Memory); you programmed it once by blowing fuses and then it held that data permanently.
I picked up a Marconi 2305 modulation meter off eBay the other day. As the name indicated, a modulation meter is used to measure the modulation characteristics of a source signal. The Marconi 2305 is capable of measuring amplitude/frequency and phase modulated signals from a few hundred kHz all the way up to 2 GHz.
The Marconi 2305 was built sometime between the late 70’s and 80’s. It is a pity that the iconic British Marconi Instruments went under in 1998 and had since changed hands a couple of times.
I just picked up a LogiMetrics A300/S 2 GHz to 4 GHz (S band) traveling wave tube amplifier (TWTA) on eBay. I had done an extreme teardown of an HP 493A TWTA a while ago and it was quite fascinating to see what’s inside of a TWT. This LogiMetrics A300/S was made from the late 70’s and unlike the HP 493A it was made entirely using solid state devices (e.g. transistors and ICs), the TWT itself of course remains a vacuum tube.
Monochromator is one of those things that has always fascinated me. Over the years, I have done quite a few experiments (I, II, III) with an EP200Mmd monochromator and it was a lot of fun. Because monochromators are such highly specialized equipment, decent ones are hard to come by at reasonable prices second hand. So my strategy has been to scour eBay once a while and pick up bit and pieces whenever I can.
Kerry Wong did a teardown of an old analog piezoelectric vibrating gyroscope:
Gyroscopes nowadays are based on micro-electro-mechanical systems (MEMS) technology. They are low cost and extremely miniaturized. A device combing both a three-axis gyroscope and a three-axis accelerometers (sometimes these devices are referred to as 6DOF devices) such as the MPU-6500 for example can be had in a QFN package as small as 3 mm x 3 mm and under 1 mm in height. Before these MEMS devices gained mainstream popularity however, larger piezoelectric vibrating gyroscopes were used in many consumer electronics devices.
After a 4 year run the dashcam on my car stopped working: a fault seems to have developed in the power system. It was mounted to the window by what I thought was just a simple mechanical mount… on further analysis it became clear that the GPS receiver was part of the mount (makes sense as the user normally glues this part to the windscreen).
Teardown and repair of an GW Instek 1080W power supply from The Signal Path:
In this episode Shahriar investigates the failure of a GW Instek 1080W power supply capable of providing up to 80V and 40A of programmable output voltage and current respectively. The power supply does not power on. However, relay noises can be heard inside the instrument during power on.
Teardown of the unit reveals a modular design with PCBs on all sides. The instrument comprises 6 different modules and 3 complete power supplies in parallel. The controller circuit is powered from the middle power supply module. Examination of the boards reveals three separate failed devices.
I have made many electronic loads in the past. For instance this simple harddrive cooler housed small dummy load, this more sophisticated constant current/constant programmable load and this heavy-duty electronic load that is capable of sinking over 1kW under peak load. In this blog post though, I am going to take a look inside an Array 3711A DC electronic load I recently purchased on eBay. You can find a video of this teardown towards the end of the post.